Skip to left navigation
Skip to page content
Laser diffraction is a widely used particle sizing technique for materials ranging from hundreds of nanometers up to several millimeters in size. The main reasons for its success are:
Laser diffraction measures particle size distributions by measuring the angular variation in intensity of light scattered as a laser beam passes through a dispersed particulate sample. Large particles scatter light at small angles relative to the laser beam and small particles scatter light at large angles, as illustrated below. The angular scattering intensity data is then analyzed to calculate the size of the particles responsible for creating the scattering pattern, using the Mie theory of light scattering. The particle size is reported as a volume equivalent sphere diameter.
Laser diffraction uses Mie theory of light scattering to calculate the particle size distribution, assuming a volume equivalent sphere model.
Mie theory requires knowledge of the optical properties (refractive index and imaginary component) of both the sample being measured, along with the refractive index of the dispersant. Usually the optical properties of the dispersant are relatively easy to find from published data, and many modern instruments will have in-built databases that include common dispersants. For samples where the optical properties are not known, the user can either measure them or estimate them using an iterative approach based upon the goodness of fit between the modeled data and the actual data collected for the sample.
A simplified approach is to use the Fraunhofer approximation, which does not require knowledge of the optical properties of the sample. This can provide accurate results for large particles. However it should be used with caution whenever working with samples which might have particles below 50µm or where the particles are relatively transparent.
Free access to exclusive content including webinars, presentations, application notes, technical notes, whitepapers and more. Enter your search term in the box and use the drop down boxes to filter the results.
Notre formation s'adresse soit à de nouveaux utilisateurs, soit à des utilisateurs souhaitant une remise à niveau ou un approfondissement de leurs connaissances ou souhaitant développer de nouvelles applications.
Aucun diplôme ...
Learn More – Discover how to get the best out of your instrument with hands-on one-on-one training sessions.
Share More – Share your Mastersizer experience with our user community.
See More – Meet technology experts and see ...
Learn More – Review theory behind laser diffraction particle sizing and how to apply it successfully to your work. Explore alternative methods like particle image analysis as well.
See More – Meet technology experts and see our lat...
This webinar will cover method development for samples dispersed in liquids. This will include a discussion of factors affecting the measurement, such as sampling and dispersion, and the relative importance of these factors in different particle size...
The end point of any particle size measurement is to provide improved product or process understanding and/or control. It order to achieve this, it is important that the size specifications selected are relevant to those products or processes. In thi...
Dry nasal powders: a new analytical challenge?
Characterizing a nasal spray formulation from droplet to API particle size
Droplet size analysis and emulsion manufacture
Laser diffraction and automated imaging: Complementary techniques for dry powder inhaler development
A novel method for measuring particle size in semisolid drug formulation
(Classroom based training)
(Webinar - Recorded)
All information supplied within is correct at time of publication.
Malvern Instruments pursues a policy of continual improvement due to technical development. We therefore reserve the right to deviate from information, descriptions, and specifications in this publication without notice. Malvern Instruments shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance or use of this material.
Malvern and the 'hills' logo, Bohlin, Gemini, Insitec, ISys, Kinexus, Mastersizer, Morphologi, Rosand, SyNIRgi, Viscotek and Zetasizer, are International Trade Marks owned by Malvern Instruments Ltd.