Skip to left navigation
Skip to page content
Laser diffraction is a widely used particle sizing technique for materials ranging from hundreds of nanometers up to several millimeters in size. The main reasons for its success are:
Laser diffraction measures particle size distributions by measuring the angular variation in intensity of light scattered as a laser beam passes through a dispersed particulate sample. Large particles scatter light at small angles relative to the laser beam and small particles scatter light at large angles, as illustrated below. The angular scattering intensity data is then analyzed to calculate the size of the particles responsible for creating the scattering pattern, using the Mie theory of light scattering. The particle size is reported as a volume equivalent sphere diameter.
Laser diffraction uses Mie theory of light scattering to calculate the particle size distribution, assuming a volume equivalent sphere model.
Mie theory requires knowledge of the optical properties (refractive index and imaginary component) of both the sample being measured, along with the refractive index of the dispersant. Usually the optical properties of the dispersant are relatively easy to find from published data, and many modern instruments will have in-built databases that include common dispersants. For samples where the optical properties are not known, the user can either measure them or estimate them using an iterative approach based upon the goodness of fit between the modeled data and the actual data collected for the sample.
A simplified approach is to use the Fraunhofer approximation, which does not require knowledge of the optical properties of the sample. This can provide accurate results for large particles. However it should be used with caution whenever working with samples which might have particles below 50µm or where the particles are relatively transparent.
Free access to exclusive content including webinars, presentations, application notes, technical notes, whitepapers and more. Enter your search term in the box and use the drop down boxes to filter the results.
The ability to quickly and reproducibly measure coffee ground particle size and size distributions is important during grinding operations to ensure that the coffee quality is maintained and that the correct grade is produced. The Mastersizer 3000 an...
Analyzing powders as a dry dispersion is in many ways simpler and faster than wet dispersion. In order to characterize a wide variety of materials accurately, it is important to control the aerodynamic dispersion. From limited quantities of Active Ph...
This recorded demo is designed to provide you with an introduction to the Mastersizer 3000. You will be able to take a look at the Mastersizer 3000 and its accessories via the webcam in our lab, and we will share the Mastersizer 3000 software with yo...
Efficient dispersion of your sample enables accurate particle sizing. But how can you tell if you are merely dispersing agglomerates or starting to break up or mill your particles? The Hydro Sight accessory for the Mastersizer range shows you all y...
This live online demo is designed to provide you with an introduction to the Mastersizer 3000. You will be able to take a look at the Mastersizer 3000 and it’s accessories via the webcam in our lab, and we will share the Mastersizer 3000 software wit...
(Webinar - Recorded)
All information supplied within is correct at time of publication.
Malvern Instruments pursues a policy of continual improvement due to technical development. We therefore reserve the right to deviate from information, descriptions, and specifications in this publication without notice. Malvern Instruments shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance or use of this material.
Malvern and the 'hills' logo, Bohlin, Gemini, Insitec, ISys, Kinexus, Mastersizer, Morphologi, Rosand, SyNIRgi, Viscotek and Zetasizer, are International Trade Marks owned by Malvern Instruments Ltd.