Skip to left navigation
Skip to page content
Laser diffraction is a widely used particle sizing technique for materials ranging from hundreds of nanometers up to several millimeters in size. The main reasons for its success are:
Laser diffraction measures particle size distributions by measuring the angular variation in intensity of light scattered as a laser beam passes through a dispersed particulate sample. Large particles scatter light at small angles relative to the laser beam and small particles scatter light at large angles, as illustrated below. The angular scattering intensity data is then analyzed to calculate the size of the particles responsible for creating the scattering pattern, using the Mie theory of light scattering. The particle size is reported as a volume equivalent sphere diameter.
Laser diffraction uses Mie theory of light scattering to calculate the particle size distribution, assuming a volume equivalent sphere model.
Mie theory requires knowledge of the optical properties (refractive index and imaginary component) of both the sample being measured, along with the refractive index of the dispersant. Usually the optical properties of the dispersant are relatively easy to find from published data, and many modern instruments will have in-built databases that include common dispersants. For samples where the optical properties are not known, the user can either measure them or estimate them using an iterative approach based upon the goodness of fit between the modeled data and the actual data collected for the sample.
A simplified approach is to use the Fraunhofer approximation, which does not require knowledge of the optical properties of the sample. This can provide accurate results for large particles. However it should be used with caution whenever working with samples which might have particles below 50µm or where the particles are relatively transparent.
Free access to exclusive content including webinars, presentations, application notes, technical notes, whitepapers and more. Enter your search term in the box and use the drop down boxes to filter the results.
Demonstrating bioequivalence (BE) for complex generics such as nasal spray suspension drug products is a challenging task. Sponsors not only have to consider device requirements but also understand the properties of the API in the presence of functio...
Spray drying, which produces a dry powder from a liquid or slurry, is one of the most popularly-used continuous drying processes in which a liquid in the form of an emulsion, wet slurry or suspension is sprayed via a spray nozzle or a rotary atomizer...
Chocolate is an affordable yet luxurious treat with mature markets in Europe and North America and developing markets around the world. The secret of chocolate’s almost universal appeal is in the mouthfeel – the “snap” of the first bite, how the choc...
This online demo is designed to get you close up and personal with the Mastersizer 3000. You will be given an insider's look at the instrument and its accessories via our lab webcam, and follow the measurement of typical pharmaceutical samples on the...
Difracción láser es una técnica ampliamente utilizada para la medición de tamaño de partícula en materiales que varían entre cientos de nanómetros hasta varios milímetros en tamaño. Es una de las tecnologías más fáciles de entender y obtener resulta...
(Webinar - Recorded)
The software really is the shining star of the Mastersizer 3000. It makes it feel radically different from other systems on the market and a really great instrument to use. I especially like the live trends which make it so quick and easy to compare data sets.
NSL Analytical Services
Throughout the research and purchasing process, we assessed at least five to six different systems before selecting the Mastersizer 3000.
Malvern came recommended by colleagues in the industry but it was the features of the Mastersizer 3000 itself that sold us on the instrument. We’ve been running for a year now without any reliability issues and with minimal running costs. Most importantly we produce data that can be trusted.
All information supplied within is correct at time of publication.
Malvern Instruments pursues a policy of continual improvement due to technical development. We therefore reserve the right to deviate from information, descriptions, and specifications in this publication without notice. Malvern Instruments shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance or use of this material.
Malvern and the 'hills' logo, Bohlin, Gemini, Insitec, ISys, Kinexus, Mastersizer, Morphologi, Rosand, SyNIRgi, Viscotek and Zetasizer, are International Trade Marks owned by Malvern Instruments Ltd.