Skip to left navigation
Skip to page content
Laser diffraction is a widely used particle sizing technique for materials ranging from hundreds of nanometers up to several millimeters in size. The main reasons for its success are:
Laser diffraction measures particle size distributions by measuring the angular variation in intensity of light scattered as a laser beam passes through a dispersed particulate sample. Large particles scatter light at small angles relative to the laser beam and small particles scatter light at large angles, as illustrated below. The angular scattering intensity data is then analyzed to calculate the size of the particles responsible for creating the scattering pattern, using the Mie theory of light scattering. The particle size is reported as a volume equivalent sphere diameter.
Laser diffraction uses Mie theory of light scattering to calculate the particle size distribution, assuming a volume equivalent sphere model.
Mie theory requires knowledge of the optical properties (refractive index and imaginary component) of both the sample being measured, along with the refractive index of the dispersant. Usually the optical properties of the dispersant are relatively easy to find from published data, and many modern instruments will have in-built databases that include common dispersants. For samples where the optical properties are not known, the user can either measure them or estimate them using an iterative approach based upon the goodness of fit between the modeled data and the actual data collected for the sample.
A simplified approach is to use the Fraunhofer approximation, which does not require knowledge of the optical properties of the sample. This can provide accurate results for large particles. However it should be used with caution whenever working with samples which might have particles below 50µm or where the particles are relatively transparent.
Free access to exclusive content including webinars, presentations, application notes, technical notes, whitepapers and more. Enter your search term in the box and use the drop down boxes to filter the results.
Este seminario ofrecerá una introducción general a la medición del tamaño de partícula mediante difracción láser. Vamos a introducir la técnica de difracción láser, así como algunos de los principios básicos de la granulometría.
Esto incluirá una v...
High alcohol based aniseed drinks (e.g. Absinthe, Ouzo, Raki, Sambuca) form a milky emulsion when water is added to them - the so-called Ouzo (or louche) effect. Adding alcohol to this emulsion recovers the clear solution. The size distribution is in...
In the second webinar in the particle size masterclass series, we will follow on from why you need to measure particle size, covered in the first masterclass, by looking at how to measure particle size. We’ll start with considerations for choosing a ...
In some industries, for example, in early-stage pharmaceutical development, it is desirable to measure particle size from a small samples of dry powders. However measuring small quantities risks measuring a sample that is not representative of the bu...
In some industries, for example, in early-stage pharmaceutical development, it is desirable to measure particle size from as little as 5mg of dry material. In this application note, we demonstrate the capability of the Aero S dry dispersion unit and ...
(Webinar - Recorded)
Particle Size Masterclass 2: which technique is best for me?
(March 19 2015)
All information supplied within is correct at time of publication.
Malvern Instruments pursues a policy of continual improvement due to technical development. We therefore reserve the right to deviate from information, descriptions, and specifications in this publication without notice. Malvern Instruments shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance or use of this material.
Malvern and the 'hills' logo, Bohlin, Gemini, Insitec, ISys, Kinexus, Mastersizer, Morphologi, Rosand, SyNIRgi, Viscotek and Zetasizer, are International Trade Marks owned by Malvern Instruments Ltd.