Skip to left navigation
Skip to page content
Static light scattering (SLS) is a technique to measure absolute molecular weight using the relationship between the intensity of light scattered by a molecule and its molecular weight and size, as described by the Rayleigh theory. In the simplest terms, Rayleigh theory says that larger molecules scatter more light than smaller molecules from a given light source and that the intensity of the scattered light is proportional to the molecule’s molecular weight.
There are two ways to measure absolute molecular weight by SLS:
Batch measurement with cuvette based instruments, such as the Zetasizer series, is an ensemble technique. Therefore the result calculated is the weight average molecular weight of the entire sample measured.
However, the most common way of measuring absolute molecular weight is to add an SLS detector e.g. Low Angle Light Scattering LALS, Right Angle Light Scattering RALS or Multi Angle Light Scattering MALS to a GPC/SEC system. By combining SLS with the separation technique you can calculate the absolute molecular weight at any point in the eluting chromatogram and determining the molecular weight of any population in a mixed sample becomes possible.
Free access to exclusive content including webinars, presentations, application notes, technical notes, whitepapers and more. Enter your search term in the box and use the drop down boxes to filter the results.
How can I trust my data? Is it affected in any way by the measurement technology, or by the user through sample preparation or the analysis parameters employed? Independent and orthogonal measurement modalities should be used to provide measurement ...
This technical note details information about maintaining Viscotek GPC/SEC systems to maximize reliability, prevent damage, reduce downtime, recognise imminent issues and minimise service visits
Triple detection GPC/SEC combines measurements from multiple detectors to offer not only increased amounts of data, but also information, which is available thanks to the combination of different detectors and would not be obtainable using the indivi...
A detailed understanding of the mechanism of unfolding, aggregation, and associated rheological changes is developed in this study for β-Lactoglobulin at different pH values through concomitant measurements utilizing dynamic light scattering (DLS), o...
Two antibody samples were characterized using column calibration and multi-detection SEC, and the results compared to show how multi-detection SEC provides more accurate data and a more complete characterization of the protein mixtures under study.
All information supplied within is correct at time of publication.
Malvern Instruments pursues a policy of continual improvement due to technical development. We therefore reserve the right to deviate from information, descriptions, and specifications in this publication without notice. Malvern Instruments shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance or use of this material.
Malvern and the 'hills' logo, Bohlin, Gemini, Insitec, ISys, Kinexus, Mastersizer, Morphologi, Rosand, SyNIRgi, Viscotek and Zetasizer, are International Trade Marks owned by Malvern Instruments Ltd.